## Friday, January 11, 2013

### Common HPLC Calculations:

Capacity Factor / Retention Factor / Capacity Ratio:  k1 (K Prime)

k1 = T(R) - T(0) / T(0)
where T(R) equals the retention time of the peak in minutes and T(0) is
the retention time of an unretained peak. *For chromatography to take place, K Prime must be > 1.00 and for most modes of chromatography, should be greater than 1.5 or 2.0 for all samples !

Tailing Factor: USP: 't'

t = W(5.0)/tw/2

where tw equals the distance between peak front and T(R) at 5% of peak height units. W(5.0) equals peak width at 5% height, in minutes.

Theoretical Plates: USP and ASTM, 'N'

N = 5.54 x (T(R)/W(50))2

Assumes width at peak half height (50)

Resolution: USP, ASTM, 'R'

R = (T(R)(b)-T(R)(a)) x 2.35/(W(50)(b) + W(50)(a))/2

Assumes width at half height (50%) with peaks (a) and (b).

*Notes: Visually, "Baseline" resolution is R = 1.5. Your goal should be R = or > 2.0. ** R of 1.5 provides 99.8% separation which means you cannot accurately quantify a 0.1% impurity so develop the method to have a resolution value of at least 2.0.

Note: The appropriate formula(s) for use with your samples may depend on which of the many pharmaceutical guidelines and regulations apply in your country. Always consult the appropriate guidelines.

1. 2. 3. 